- switch on the computer
- on the NanoZoomer main switch
- launch the NDP.scan software
- exit NDP.scan
- switch off the Nanozoomer
- switch off the computer
- if the lamp is off (no green LED above the on/off button), let the ventilator cool down the lamp unit at least 10 minutes before switch of the main switch of the lamp unit
The following schema sketches out the light path of the NanoZoomer. This is a normal fluorescent setup with the particularity of the Camera. This RGB camera always acquires 3x8bit -> 24bit RGB images, even if we scan one single fluorescence channel.
Acquisition and viewer software
More documentation about the NDP software suite is on this dedicated page.